Industrial High Voltage X Ray Control Unit
Specifications(example for High Voltage Control unit of 450KV machine):
English operation interface
PLC intelligent microcomputer operation, touching screen displaying
Filament automatic preheating
Voltage controlling and indication:50~450kV,step distance 1kV,precision ±0.5%
Repeatability:±0.005% of max voltage value (under normal temperature)
Current controlling and indication:
Big focus 1~10mA,step distance 0.1mA,precision ±0.5%
Small focus 1~2mA,step distance 0.1mA,precision ±0.5%
Tube voltage, current preset, preset value can be adjust continuously, system also have up-limit and hardware protection.
Exposure time preset and indication:0~999min random setting
Working mode selection:"RTR" and "Filming"
Indication of tube temperature
Multiple protection for on-off high voltage, match with key switch, avoid misusing of X-ray machine. It has sound reminder when open or close the voltage
Automatic and manual aging status selection, for automatic aging can set up-limit voltage and current according to using requirement.
Big/small focal working status selection, operated by key switch, and prevent destroy the X-ray tube when transmit the focus during HV on, and inside of procedure it has focus interlock protection functions.
Working time acceleration
Failure alarming, feedback reminding and (light, sound)protect functions :
Over voltage protection
Over current protection
Low current protection
Low flow of cooler protection
No voltage protection
No current protection
Tube head over temperature protection
Lead door switch interlock protection
Pic of control table:
Looking for ideal High Voltage Control Unit Manufacturer & supplier ? We have a wide selection at great prices to help you get creative. All the Radiation X Ray Detector Controller are quality guaranteed. We are China Origin Factory of X Ray Controller Adjust Current. If you have any question, please feel free to contact us.
Product Categories : Units For X Ray RTR System